Frequently Asked Questions
The low side (T1) & high side (T2) temperature clamps can be calibrated to ice water (32° F). Accuracies of one degree or better are obtained by stabilizing a source of ice water preferably in an insulated cooler. Submerge low side (T1) temperature clamp into the ice water and begin meter calibration procedure:
Low side (T1) must be saved before calibrating high side (T2).
Repeat the same procedure to calibrate high side (T2).
Pressure Calibration Information.